Read online or download PDF • Page 44 / 67 • KROHNE IFC Converter EN User Manual • KROHNE Equipment. Micro Precision provides expert calibration services for KROHNE instruments including KROHNE IFC D -. Select from options for the IFC D which. KROHNE 08/ Subject to change without notice. Applicable to. Software-Versions. ○ IFC _. / D. Display version. No. and.
|Published (Last):||7 December 2012|
|PDF File Size:||15.87 Mb|
|ePub File Size:||15.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
Passive mode External power source required for operation of the outputs.
What’s missing? Tell us about it.
V ariable area flowmeters. Or point us to the URL where the manual is located. P Pulse output overranged. Tell us about it. All operating data have been factory set in accordance with your specifications.
KROHNE IFC 010 Converter Short EN User Manual
Setting ranges The ranges are dependent on the meter size DN and the flow velocity v: Do not cross or loop cables inside the terminal compartment. Use separate cable entries see below for power supply, field current cables, signal lines, outputs and inputs. Page 10 3 Start-up 3.
Your manual failed to upload Page 11 Standard factory f Function 1. Page 14 Function 1. Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
KROHNE IFC Converter Short EN User Manual | 19 pages
These concise instructions do not include the following: Tell us what’s missing. Switches, counters, displa ys and recorders.
Power the unit, and the flowmeter will immediately start process flow measurement. Don’t see a manual you are looking for?
Signal cable to be permanently laid. Kdohne circuits and output circuits are designed to meet the standards of overvoltage classes III and II, respectively.
Connect shields via stranded drain wires. Vo r tex flowmeters. Page 15 Function 3.
Page 2 Contents 1 1. Underwater and underground installation possible.
If installed and operated in accordance with these operating instructions, it will rarely present any problems. Lev el measuring instruments. The selected creepage distances and clearances have been sized in conformity with VDE and IEC for contamination category 2. Page 8 Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
Page 3 1 Electrical connection: The operator is, howeverobligated to take note of these sections in the. The flowmeter housings protecting the electronic equipment from dust and moisture must always be kept closed.